IEC60068 50J Vertical Hammer Impact Testing Machine
- PRODUCT DETAIL
|Usage:||Impact Testing Machine||Standard:||IEC60068-2-75 Test Ehc|
|Height Distance:||0-1000mm||Hammer:||Optional 1J/2J/5J/10J/20J/50J|
|High Light:||50J Impact Testing Machine , IEC60068 Impact Testing Machine , 20J impact testing equipment|
IEC60068 Test Ehc Vertical Hammer Impact Test Apparatus Easy Operate
The vertical hammer impact test apparatus conforms to the standard requirement of the IEC60068-2-75 Test Ehc. The hammer consists basically of a striking element which falls freely from rest through a vertical height, selected from following table, on to the specimen surface held in a horizontal plane. The fall of the striking element shall be along a guideway, for example a tube, with negligible braking. This guideway shall not rest on the specimen and the striking element shall be free of the guideway on striking the specimen.
The height distance:0-1000mm
The hammer: 2J, stainless steel
The conduit: Φ35×300mm
The way of releasing: by manual
Table-Height of fall:
|Equivalent Mass kg||0.25||0.25||0.25||0.25||0.25||0.5||1.7||5||5||10|
|Height of fall mm±1%||80||140||200||280||400||400||300||200||400||500|
The height of fall shall be as giver in above table, the equivalent mass stated therein being equal to the actual mass of the striking element.
A proper hammer is chose according to the related standard requirement. Place the sample on the bottom of the conduit and adjust the height of the conduit the height is measured by the ruler) Next, put the hammer on the top of the conduit and press the release button. Thus the impact hammer falls down and hit the sample.